Power Analyzers
Sponsor Your Catalog 

FU2040  Panel Power Meter Introduction
FU2040 is a LCD display power analyzer, metering and displaying all the measured parameter in one circuit and computing four quadrants energy. This model can work in wide temperature range.
Measurement Parameters
Voltage (PhN); Voltage (PhPh); Current; Frequency; PF; Active Power(W); Reactive Power(Q) ; Apparent Power(S)
Computation
Forward / Reverse active / reactive power energy.
Measuring Range
30～600V, 0～6A, 45～60Hz
Measuring Accuracy
Frequency: 0.1%
Electric energy: 0.5%, 1.0%
Voltage : 0.2%±0.1V
Current : 0.2%±0.001A
Power : 0.5% ±0.4W
Power Factor : 0.5% ±0.001Beijing GFUVE Electronics 
GF3021  Panel Meter Calibrator The GF3021 Panel Meter Calibrator (PMC) integrates eight calibration functions: voltage, current, frequency, power, power factor, phase, reactive power (VARs), and synchroscope into one
compact, portable package. All of these functions in one lightweight yet rugged instrument, allow you to easily calibrate virtually every type of panel meter and many types of transducers, circuit breakers and overcurrent relays. A handheld control is also provided, so you can operate theGF3021 at a distance when calibrating panel meters in the control room or substation.
The GF3021's precision ac/dc voltage and current
sources may operate independently to calibrate voltmeters,
ammeters and frequency meters, or together to calibrate
watt meters, watthour meters, power factor meters, VAR
meters, and phase meters.
The GF3021 is also an excellent source for
calibrating highaccuracy transducers and revenue meters,
when used with an external standard. The GF302
Power System Analyzer is a good companion instrument
for these tests, with basic accuracy of 0.05% and 0.05°,
builtin transducer output measurement capability and
timer. The Model 931A also performs a wide variety of
other measurements. Together, these two instruments
can calibrate and measure the vast majority of your work
load requirements.Beijing GFUVE Electronics 
FU2000  Multifunction Internet Power Analyzer Introduction
FU2000 is a multifunction power analyzer, It can measure all the electric parameters in the circuit, and it has RTC, calendar and one pulse output.
Measurement Parameters
Voltage (PhN); Voltage (PhPh); Current; Frequency; PF; Active Power (W); Reactive Power (Q); Apparent Power(S)
Computation
Forward active power energy
Reverse active power energy
Forward active power energy
Reverse reactive power energy
Measuring Range
30～600V, 0～6A, 45～65Hz, 1~0~1
Measuring Accuracy
Frequency: 0.1%
Electric energy: 0.5%, 1.0%
Voltage : 0.2%±0.1V
Current : 0.2%±0.001A
Power : 0.5% ±0.4W
Power Factor : 0.5% ±0.001Beijing GFUVE Electronics 
382100  1200A 3Phase Power Analyzer/Datalogger Datalogging Power Analyzer (up to 30,000 sets of measurements). Single phase, 3phase/3wire or 3phase/4wire systems. Large backlighting LCD displays 35 parameters simultaneously (MW, kVAR, MVA, PF plus the voltage and current of the 3 phases). Displays Active Power (kW), Apparent Power (kVA), Reactive Power (kVAR) and Power Factor. Current measurements to 1200A and Voltage measurements to 600V. MWH, MVAh, MVARh energy measurements. Adjustable CT ratio (1 to 600) and PT ratio (1 to 1000). Automatically logs up to 30,000 sets of measurements with 2GB SD memory card. Userprogrammable sampling time from 2 to 7200 seconds.Extech Instruments

DMII PLUS  1000A Power Quality Analyzer The Amprobe DMII PLUS is a fullfeatured three phase power quality recorder used to maintain, diagnose andtroubleshoot electrical systems. It measures and records a broad spectrum of power quality parameters. Easy to use, it’s capable of measuring and recording voltage, current, voltage anomalies (sags or surges), harmonics, energy and power (active, reactive and apparent), power factor (true and displacement), and frequency.Amprobe Test Tools

380820  Universal AC Power Source & AC Power Analyzer Standby Power Measurement (10mW to 100W) for desktop and notebook PCs, LCD TVs, AC adaptor/chargers
For testing products from 85 to 250VAC, 45 to 65Hz
Standard Power Measurement:
100mW to 1000W (110V)
100mW to 2000W (220V)
Power Quality Measurement (Watts, Power Factor, Voltage, Current, Harmonics)
Cost Function calculates cost per day, month, and year
99 memory locations for preset voltage and frequency output settings
Userprogrammable Over Current Protection
Programmable through the builtin RS232 interfaceExtech Instruments 
PROVA6200  Graphic Power Quality Analyzer Analysis for Single Phase and Balanced 3 phase system. True RMS Value(V and I). Active Power(K,KW,MW,GW). Apparent(VA,KVA,MVA) and Reactive Power (VAR,KVAR,MVAR). Power Factor(PF),Phase Angle( £X ). Energy(WH,KWH,KVARH,PFH). Programmable PT(1 to 3000)RatiosTES Electrical Electronic

HA1600A  Mains & Harmonics Analyser with Flicker Meter Compliance quality measurements to EN6100032 and EN6100033. Measures peak or rms voltage or current, real or apparent power, power factor, phase etc. Tabular/histogram of 40 harmonics.Thurlby Thandar Instruments

DW6090  POWER ANALYZER POWER ANALYZER  Model : DW6090 * Large LCD, multidisplays. * True rms for WATT, ACV, ACA. * Watt, VA, Power factor, ACV, ACA, VAr, Whr, DCV, DCA, ohms, RS232. * Accept current probe & CT input. Battery power.Lutron Electronic

DW6091  POWER ANALYZER POWER ANALYZER  Model : DW6091 * Large LCD, multidisplays. * True rms Watt, VA, Whr, power factor Hz, ACV, ACA, RS232 interface. * Accept external CT input. Battery power.Lutron Electronic

2335A  Wideband Power Analyzer The Model 2335A Wideband Power Analyzer is a precision, high accuracy, high frequency autoranging instrument which measures and displays true rms Voltage, true rms Current, and true mean Power over a frequency range from dc to more than 1MHz. Full scale Current and Voltage inputs are typically measured within �0.1% of the amplitude reading to at least 500kHz. The corresponding Power is typically measured to within �0.1% of the input VoltAmperes to 250kHz and to within �0.2% of the input VoltAmperes to 500kHz for loads having any Power Factor.ClarkeHess Communications

PCEPA 8000  threephase power analyzer The PCEPA 8000 power analyzer is able to measure and record the power in both singlephase or threephase circuits. With the PCEPA 8000 power analyzer you will be able to perform prolonged measurements. Measurement values are saved in xls format in the SD card of our power analyzer in order to be easily analyzed on the computer. The measurement rate can be set from 2 to 7200 seconds. The PCEPA 8000 power analyzer can also measure power and determine both the power factor and the phase angle. The clamps of the PCEPA 8000 power analyzer can be used for cables with a diameter of 50 mm. The PCEPA 8000 is the ideal tool to be used in power supply networks. Its 3.7" display allows the user to read all measured values for a quick analysis and also provides a view of the current, voltage and power. Here you will find an overview of the power analyzers that best suit your needs.PCE Instruments

PS3000  Energy Analyzer. The PS3000 provides simplicity and reliability in the analysis of electric power. Complete power/energy/cost analysis. Measures voltage, current, KW, KVA, KVAR, true power factor, displacement power factor, KWH, elapsed and estimated cost, and duty cycle. Complete logging and manual waveform capture ability.Summit Technology

2510  Power analyzer with RS232 C interface Multifunction power analyzer for testing of Watt, Power factor, ACV, ACA, DCA Hz and Ohm. Super large LCD 93 x 52 mm, easy to read out the watt, power factor, voltage and current value at the same time.PeakTech Prüf Und

1000A  3Phase Power Analyzer/Datalogger Large backlighting LCD displays 10 parameters simultaneously (kW, kVAR, kVA, PF plus the voltage and current of the the 3 phases) Up to four current probes can be connected to the analyzer Clampon True RMS power measurements with softwares for Harmonics analysis Displays Active Power (kW), Apparent Power (kVA), Reactive Power (kVAR) and Power Factor Current measurements to 1000A and Voltage measurements to 600V kWH, kVARh and kVAh energy measurements Automatically logs up to 20,000 sets of measurements and manually logs up to 99 setsAYAInstruments

PS3000 Energy Analyzer  Energy Analyzer Complete power/energy/cost analysis. Measures voltage, current KW, KVA, KVAR, true power factor, displacement power factor, KWH, elapsed and estimated cost, and duty cycle. Complete logging and manual waveform capture ability.AYAInstruments

MAP  MV/LV Electrical Network Analyzers A global solution for monitoring electrical networks, stations and HVA/HVB installations. Slotin or rack MAP analyzers measure all MV/LV electrical network parameters: RMS voltage and current, active, reactive, apparent and distorting powers, energies, cos phi, power factorEnerdis

PROVA6800  Power Harmonics Analyzer Analysis for 3P4W, 3P3W, 1P2W, 1P3W, True RMS Value(V 123 and L 123 ), Active Power(W, KW, MW, GW), Apparent and Reactive Power (KVA, KVAR), Power Factor (PF), Phase Angle (F), Energy (WH, KWH, KVARH, PFH)TES Electrical Electronic

PS3500  Power Analyzer with Bluetooth and removable SD memory card slot OUR MOST POPULAR ENERGY ANALYZER REINVENTED! Get complete power/energy/cost analysis in a compact, rugged package. The PS3500 measures voltage, current, KW, KVA, KVAR, true power factor, displacement power factor, KWh, Hz, elapsed and estimated cost, duty cycle. Long term logging for up to months at a time.Summit Technology

DPATT3B  Three Phase Power Analyser Parameters : Voltage ( V)  V1,V2,V3 & Avg. Voltage Current (A) : I1 , I2 , I3 & Avg. Current System Frequency : (Hz) Active Power(W) : W1,W2,W3 & Total Wattage Power Factor ( Cosø): PF1, PF2, PF3 & System Apparent Power ( VA): Total Measurement : True RMS Phase Voltage : 3  600 V / 10  1200V AC Phase Current : 50mA  80A AC System Frequency : 40 to 70 Hz For Power Factor : 0.05 Lead to unity to 0.05 LagUMA Electronics Enterprises

CTERP2000  Relaying Current Transformer Analyzer Measures ratio, CT polarity, winding resistance, remanance and insulation resistance of Relaying Current Transformers. Measures the hysteresis losses of CT core and the power factor of hysteresis losses.Eltel Industries

NET VISION 2010  THREEPHASE NETWORK ANALYZER NET VISION 2010 is a portable device designed for a quick and precise measurement of the electric values of a threephase network. It is used by power engineers for check of AC motors, relay current protections, electricity meters, etc. The true RMS values of the threephases of voltage and one current are measured simultaneously. Active power, reactive power, power factor, sequence of rotation, frequency, phase angle between two voltages, phase angle between voltage and current, peak values, crest factor are computed and indicated.EMSYST Ltd

3196  Power Quality AnalyzerHTI Single Phase & Three Phase. Color LCD Display. Measure power factor. Sags. Swells. Flicker. Transients 2MHz (500nanosec). Harmonics to the 50th. Detection and waveform display. Four current and Four voltage channelsHIOKI E.E.

3197  POWER QUALITY ANALYZER Catch power quality problems on the fly, before they catch you! Monitor for : Inrush current, Voltage swell, Voltage dips, Transient overvoltage, Interruptions Measure and record : Power and Power factor, Active/Reactive energy, Demand, Load changes (with graph display), Voltage and currentHIOKI E.E.
Results 1  24 of 34
Power Analyzers  Measure the quantity, quality, and frequency of electric power.
Search Filters
 Power Factor x
Types
Features
 AC (9)
 AC Power Sources (5)
 Analysis (10)
 Current (26)
 Energy (12)
 Energy Analyzers (12)
 Frequency (11)
 Harmonic (9)
 Interfaces (6)
 kVA (7)
 kW (7)
 kWh (6)
 Measurement (11)
 Phase (17)
 Phase Angle (6)
 Power Quality (6)
 Single Phase (8)
 Sources (6)
 Three Phase (11)
 Voltage (20)
 »See All 161 Features